Water potential is the most fundamental and essential measurement in soil physics because it describes the force that drives water movement.
Water potential helps researchers determine how much water is available to plants.
Making good water potential measurements is largely a function of choosing the right instrument and using it skillfully. In an ideal world, there would be one instrument that simply and accurately measured water potential over its entire range from wet to dry. In the real world, there is an assortment of instruments, each with its unique personality. Each has its quirks, advantages, and disadvantages. Each has a well-defined range.
Below is a comparison of water potential instruments and the ranges they measure.
A comparison of water potential instrument ranges
To learn more about measuring water potential, see the articles or videos below:
Soil moisture release curves have always had two weak areas: a span of limited data between 0 and -100 kPa and a gap around field capacity where no instrument could make accurate measurements.
Using HYPROP with the redesigned WP4C, a skilled experimenter can now make complete high resolution moisture release curves.
Between 0 and -100 kPa, soil loses half or more of its water content. If you use pressure plates to create data points for this section of a soil moisture release curve, the curve will be based on only five data points.
And then there’s the gap. The lowest tensiometer readings cut out at -0.85 MPa, while historically the highest WP4 water potential meter range barely reached -1 MPa. That left a hole in the curve right in the middle of plant-available range.
The conversion of light energy and atmospheric carbon dioxide to plant biomass is fundamentally important to both agricultural and natural ecosystems.
The detailed biophysical and biochemical processes by which this occurs are well understood. At a less-detailed level, however, it is often useful to have a simple model that can be used to understand and analyze parts of an ecosystem. Such a model has been provided by Monteith (1977). He observed that when biomass accumulation by a plant community is plotted as a function of the accumulated solar radiation intercepted by the community, the result is a straight line. Figure 1 shows Monteith’s results.
Figure 1. Total dry matter produced by a crop as a function of total intercepted radiation (from Monteith, 1977).
The PHYTOS 31 Leaf Wetness Sensor was designed to measure the presence and duration of water on leaf surfaces. However, Dr. Bruce Bugbee, professor of Crop Physiology at Utah State University, noticed that his leaf wetness sensor revealed interesting phenomena associated with some precipitation events. Here is what he observed on a recent day at the USU Environmental Observatory in Logan, Utah
It is possible to have a day with numerous 0.1 mm increments of rain, followed by some evaporation, in which a rain gauge would not record any rain during the day.
“Recent data from our weather station provided two examples of the offset in measurement associated with tipping bucket rain gauges. It started raining on campus last night at exactly 20:00 hours, as indicated by the response of the leaf wetness sensor (Figure 1). The first 0.1 mm tip of the rain gauge occurred about 25 minutes later (Figure 2). The resolution for most high-quality tipping bucket rain gauges is listed as 0.1 mm, but this is not the resolution for the first 0.1 mm of rain.
We occasionally see soil moisture sensors damaged by lightning. Here’s what to do to protect them.
The secondary products of a lightning strike include electromagnetic pulses, electrostatic pulses, and earth current transients.
Surge suppression components typically perform their suppression function by temporarily short circuiting the voltage between two wires, several devices, or ground.
Electromagnetic pulses are created by the strong magnetic field that is formed by the short term current flow taking place in the lightning strike. With current flows as high as 510kA per microsecond, these currents create very large magnetic fields. These short term magnetic fields then induce voltages onto wires and cables.
Electrostatic pulses are created by electrostatic fields that accompany a thunderstorm. Any cable suspended above the earth during a thunderstorm is immersed in the electrostatic field and will be electrically charged. Quick changes in the charges stored in both the clouds and earth take place whenever there is a lightning strike. The charge on the cable must now be discharged or neutralized. Unable to find a path to ground (earth), it breaks down insulation and component in its efforts to return to earth.
Earth current transients are the direct result of the neutralization process that immediately follows the end of lightning strike. Neutralization is accomplished by the movement or redistribution of charge along or near the earth’s surface from all the points where the charge had been initially induced to the point where the lightning strike has just terminated. Earth current transients create a shift in potential across a ground plan, often called a “ground bounce”.
Get better air temperature accuracy with this new method
Accurate air temperature is crucial for microclimate monitoring
The accuracy of air temperature measurement in microclimate monitoring is crucial because the quality of so many other measurements depend on it. But accurate air temperature is more complicated than it looks, and higher accuracy costs money. Most people know if you expose an air temperature sensor to the sun, the resulting radiative heating will introduce large errors. So how can the economical ATMOS 41’s new, non-radiation-shielded air temperature sensor technology be more accurate than typical radiation-shielded sensors?
We performed a series of tests to see how the ATMOS 41’s air temperature measurement compared to other sensors, and the results were surprising, even to us. Learn the results of our experiments and the new science behind the extraordinary accuracy of the ATMOS 41’s breakthrough air temperature sensor technology.
In this brief 30-minute webinar, find out:
Why you should care about air temperature accuracy
Where errors in air temperature measurement originate
The first principles energy balance equation and why it matters
Results of experiments comparing shielded sensor accuracy against the ATMOS 41
The science behind the ATMOS 41 and why its unshielded measurement actually works
Patterns of water replenishment and use give rise to large spatial variations in soil moisture over the depth of the soil profile. Accurate measurements of profile water content are therefore the basis of any water budget study. When monitored accurately, profile measurements show the rates of water use, amounts of deep percolation, and amounts of water stored for plant use.
Three common challenges to making high-quality volumetric water content measurements are:
making sure the probe is installed in undisturbed soil,
minimizing disturbance to roots and biopores in the measurement volume, and
eliminating preferential water flow to, and around, the probe.
All dielectric probes are most sensitive at the surface of the probe. Any loss of contact between the probe and the soil or compaction of soil at the probe surface can result in large measurement errors. Water ponding on the surface and running in preferential paths down probe installation holes can also cause large measurement errors.
Installing soil moisture sensors will always involve some digging. How do you accurately sample the profile while disturbing the soil as little as possible? Consider the pros and cons of five different profile sampling strategies.
Preferential flow is a common issue with commercial profile probes
Profile probes are a one-stop solution for profile water content measurements. One probe installed in a single hole can give readings at many depths. Profile probes can work well, but proper installation can be tricky, and the tolerances are tight. It’s hard to drill a single, deep hole precisely enough to ensure contact along the entire surface of the probe. Backfilling to improve contact results in repacking and measurement errors. The profile probe is also especially susceptible to preferential-flow problems down the long surface of the access tube.
Trench installation is arduous
Installing sensors at different depths through the side wall of a trench is an easy and precise method, but the actual digging of the trench is a lot of work. This method puts the probes in undisturbed soil without packing or preferential water-flow problems, but because it involves excavation, it’s typically only used when the trench is dug for other reasons or when the soil is so stony or full of gravel that no other method will work. The excavated area should be filled and repacked to about the same density as the original soil to avoid undue edge effects.
Digging a trench is a lot of work.
Augur side-wall installation is less work
Installing probes through the side wall of a single augur hole has many of the advantages of the trench method without the heavy equipment. This method was used by Bogena et al. with EC-5 probes. They made an apparatus to install probes at several depths simultaneously. As with trench installation, the hole should be filled and repacked to approximately the pre-sampling density to avoid edge effects.
Multiple-hole installation protects against failures
Digging a separate access hole for each depth ensures that each probe is installed into undisturbed soil at the bottom of its own hole. As with all methods, take care to assure that there is no preferential water flow into the refilled augur holes, but a failure on a single hole doesn’t jeopardize all the data, as it would if all the measurements were made in a single hole.
The main drawback to this method is that a hole must be dug for each depth in the profile. The holes are small, however, so they are usually easy to dig.
Single-hole installation is least desirable
It is possible to measure profile moisture by auguring a single hole, installing one sensor at the bottom, then repacking the hole, while installing sensors into the repacked soil at the desired depths as you go. However, because the repacked soil can have a different bulk density than it had in its undisturbed state and because the profile has been completely altered as the soil is excavated, mixed, and repacked, this is the least desirable of the methods discussed. Still, single-hole installation may be entirely satisfactory for some purposes. If the installation is allowed to equilibrate with the surrounding soil and roots are allowed to grow into the soil, relative changes in the disturbed soil should mirror those in the surroundings.
Bogena, H. R., A. Weuthen, U. Rosenbaum, J. A. Huisman, and H. Vereecken. “SoilNet-A Zigbee-based soil moisture sensor network.” In AGU Fall Meeting Abstracts. 2007. Article link.
This week, we continue highlighting the second of two current research projects (see part one) which use soil moisture sensors to measure volumetric water content in tree stems and why this previously difficult to obtain measurement will change how we look at tree water use.
Tamarisk tree: an invasive species dominant in Sudan and arid parts of the United States. (Photo credit: biolib.cz)
Determining Tree Stem Water Content in Drought Tolerant Species
Tadaomi Saito and his research team were interested in using dielectric soil moisture sensors to measure the tree stem volumetric water content of mesquite trees and tamarisk, two invasive species dominant in Sudan and arid parts of the United States. Mesquite is a species that can access deep groundwater sources using their taproots which is how they compete with native species. Tamarisk, on the other hand, uses shallow, saline groundwater to survive. The team wanted to see if dielectric probes were useful for real-time measurement of plant water stress in these drought tolerant species and if these measurements could illuminate differing tree water-use patterns. These sensors could then potentially be used for precision irrigation strategies to assist in agricultural water management.
Temperature Calibration Was Essential
After calibrating the soil moisture sensors to the wood types in a lab, the team inserted probes into the stems of both trees. They also monitored groundwater and soil moisture content to try and infer whether or not the trees were plugged into a deep source of water. Interestingly, Saito found that, unlike soil, where temperature fluctuation is buffered, tree stems are subject to large variations in temperature throughout the course of the day. This temperature fluctuation interfered with the soil moisture probes’ ability to accurately measure VWC. The team came up with a simple method for accounting for temperature variability and were then able to obtain accurate VWC measurements.
Saito’s results were similar to Ashley Matheny’s study (see part 1), in that they found a lot of different patterns, even in trees of the same species. Water-use depended on where the trees were on the landscape. Some of them were tapped into groundwater, and the stem water storage didn’t change no matter how dry the soil became. Whereas others, depending on their position in the landscape, were very dependent on soil moisture conditions.
Saito’s study illustrates that we see everything about a tree that’s above ground, but we may have no sense of what’s going on below ground. We can put a soil moisture sensor in the ground and decide there’s plenty of moisture available. Or if conditions are dry, we may decide the tree is under drought stress, but we don’t know if that tree is tapped into a more permanent source of groundwater.
Other researchers have put soil moisture sensors in orchards looking at stem water storage from a practical standpoint for irrigation management. Their data didn’t work out so well because of cable sensitivity where water on the cable created false readings. However, the data they were able to obtain showed that some of the trees were plugged into water sources that were independent of the soil. Those trees were able to withstand drought and needed less irrigation, whereas other trees were much more sensitive to soil moisture.
If we had an inexpensive, easy to deploy measurement device plugged into every tree in an orchard, we could irrigate tree by tree, give them precisely what they needed, and account for their unique situation.
What Does it All Mean?
The interesting thing about using soil moisture sensors in a tree is that stem water content is a difficult to obtain piece of information that has now been made easier. Historically, we’ve focused on measuring sap flow, but that’s just how much water is flowing past the sensor. We’ve measured what’s in the soil: a pool of moisture that’s available to the tree. But some trees are huge in size, such as ones along the coast of California. They’re able to store vast amounts of water above-ground in their tissue. Understanding how a tree can use that water to buffer or get through periods of drought is a unique research topic that has had very little attention. With these kinds of sensors, we can start to investigate those questions.
Reference: Saito T., H. Yasuda, M. Sakurai, K. Acharya, S. Sueki, K. Inosako, K. Yoda, H. Fujimaki, M. Abd Elbasit, A. Eldoma and H. Nawata , Monitoring of stem water content of native/invasive trees in arid environments using GS3 soil moisture sensor , Vadose Zone Journal , vol.15 (0) (p.1 – 9) , 2016.03
Get more information on applied environmental research in our
In an update to our previous blog, “Soil Moisture Sensors in a Tree?”, we highlight two current research projects using soil moisture sensors to measure volumetric water content (VWC) in tree stems and share why this previously difficult-to-obtain measurement will change how we look at tree water usage.
Researchers explore the feasibility of inserting capacitance soil sensors in tree stems as a real-time measurement.
Soil Moisture Sensors in Tree Stems?
In a recent research project, Ph.D. candidate Ashley Matheny of the University of Michigan used soil sensors to measure volumetric water content in the stems of two species of hardwood trees in a northern Michigan forest: mature red oak and red maple. Though both tree types are classified as deciduous, they have different strategies for how they use water. Oak is anisohydric, meaning the species doesn’t control their stomata to reduce transpiration, even in drought conditions. Isohydric maples are more conservative. If the soil starts to dry out, maple trees will maintain their leaf water potential by closing their stomata to conserve water. Ashley and her research team wanted to understand the different ways these two types of trees use stem water in various soil moisture scenarios.
Historically, tree water storage has been measured using dendrometers and sap flow data, but Ashley’s team wanted to explore the feasibility of inserting a capacitance-type soil sensor in the tree stems as a real-time measurement. They hoped for a practical way to make this measurement to provide more accurate estimations of transpiration for use in global models.
Scientists measured volumetric water content in the stems of two species of hardwood trees in a northern Michigan forest: mature red oak and red maple.
Ashley and her team used meteorological, sap flux, and stem water content measurements to test the effectiveness of capacitance sensors for measuring tree water storage and water use dynamics in one red maple and one red oak tree of similar size, height, canopy position and proximity to one another (Matheny et al. 2015). They installed both long and short soil moisture probes in the top and the bottom of the maple and oak tree stems, taking continuous measurements for two months. They calibrated the sensors to the density of the maple and oak woods and then inserted the sensors into drilled pilot holes. They also measured soil moisture and temperature for reference, eventually converting soil moisture measurements to water potential values.
Results Varied According to Species
The research team found that the VWC measurements in the stems described tree storage dynamics which correlated well with average sap flux dynamics. They observed exactly what they assumed would be the anisohydric and isohydric characteristics in both trees. When soil water decreased, they saw that red oak used up everything that was stored in the stem, even though there wasn’t much available soil moisture. Whereas in maple, the water in the stem was more closely tied to the amount of soil water. After precipitation, maple trees used the water stored in their stem and replaced it with more soil water. But, when soil moisture declined, they held onto that water and used it at a slower rate.
Researchers want to figure out the appropriate level of detail for tree water-use strategy in a global model.
Trees use different strategies at the species level
The ability to make a stem water content measurement was important to these researchers because much of their work deals with global models representing forests in the broadest sense possible. They want to figure out the appropriate level of detail for tree water-use strategy in a global model. Both oak and the maple are classified as broadleaf deciduous, and in a global model, they’re lumped into the same category. But this study illustrates that if you’re interested in hydrodynamics (the way that trees use water), deciduous trees use different strategies at the species level. Thus, there is a need to treat them differently to produce accurate models.
Reference: Matheny, A. M., G. Bohrer, S. R. Garrity, T. H. Morin, C. J. Howard, and C. S. Vogel. 2015. Observations of stem water storage in trees of opposing hydraulic strategies. Ecosphere 6(9):165. http://dx.doi.org/10.1890/ES15-00170.1
Next week: Learn about more research being done using soil moisture sensors to measure volumetric water content in tree stems.
Get more information on applied environmental research in our
Salt in soil comes from the fertilizer we apply but also from irrigation water and dissolving soil minerals. If more salt is applied in the irrigation water than is leached or taken off in harvested plants, the soil becomes more saline and eventually ceases to support agricultural production (see part 1). This week, learn an effective way to measure electrical conductivity (EC) in soil.
Salt in irrigation water reduces its water potential, making it less available to the plant.
How to Measure Electrical Conductivity of the Soil Solution
As mentioned above, the earliest measurements of solution conductivity were made on soil samples, but it was found to be more reliable to extract the soil solution and make the measurements on it. When values for unsaturated soils are needed, those are calculated based on the saturation numbers and conjecture about how the soil dried to its present state. Obviously a direct measurement of the soil solution conductivity would be better if it could be made reliably.
Two approaches have been made to this measurement. The first uses platinum electrodes embedded in ceramic with a bubbling pressure of 15 bars. Over the plant growth range the ceramic remains saturated, even though the soil is not saturated, allowing a measurement of the solution in the ceramic. As long as there is adequate exchange between the ceramic and the soil solution, this measurement will be the EC of the soil solution, pore water EC.
Salt in soil comes from the fertilizer we apply, irrigation water and dissolving soil minerals.
The other method measures the conductivity of the bulk soil and then uses empirical or theoretical equations to determine the pore water EC. The ECH2O 5TE uses the second method. It requires no exchange of salt between soil and sensor and is therefore more likely to indicate the actual solution electrical conductivity. The following analysis shows one of several methods for determining the electrical conductivity of the saturation extract from measurements of the bulk soil electrical conductivity.
Mualem and Friedman (1991) proposed a model based on soil hydraulic properties. It assumes two parallel conduction paths: one along the surface of soil particles and the other through the soil water. The model is
Here σb is the bulk conductivity which is measured by the probe, σs is the bulk surface conductivity, σw is the conductivity of the pore water, θ is the volumetric water content, θs is the saturation water content of the soil and n is an empirical parameter with a suggested value around 0.5. If, for the moment, we ignore surface conductivity, and use eq. 1 to compute the electrical conductivity of a saturated paste (assuming n = 0.5 and θs = 0.5) we obtain σb = 0.35σw. Obviously, if no soil were there, the bulk reading would equal the electrical conductivity of the water. But when soil is there, the bulk conductivity is about a third of the solution conductivity. This happens because soil particles take up some of the space, decreasing the cross section for ion flow and increasing the distance ions must travel (around particles) to move from one electrode of the probe to the other. In unsaturated soil these same concepts apply, but here both soil particles and empty pores interfere with ion transport, so the bulk conductivity becomes an even smaller fraction of pore water conductivity.
When water evaporates at the soil surface, or from leaves, it is pure, containing no salt, so evapotranspiration concentrates the salts in the soil.
Our interest, of course, is in the pore water conductivity. Inverting eq. 1 we obtain
In order to know pore water conductivity from measurements in the soil we must also know the soil water content, the saturation water content, and the surface conductivity. The 5TE measures the water content. The saturation water content can be computed from the bulk density of the soil
Where ρb is the soil bulk density and ρs is the density of the solid particles, which in mineral soils is taken to be around 2.65 Mg/m3 . The surface conductivity is assumed to be zero for coarse textured soil. Therefore, using the 5TE allows us to quantify pore water EC through the use of the above assumptions. This knowledge has the potential to be a very useful tool in fertilizer scheduling.
Electrical Conductivity is Temperature Dependent
Electrical conductivity of solutions or soils changes by about 2% per Celsius degree. Because of this, measurements must be corrected for temperature in order to be useful. Richards (1954) provides a table for correcting the readings taken at any temperature to readings at 25 °C. The following polynomial summarizes the table
where t is the Celsius temperature. This equation is programmed into the 5TE, so temperature corrections are automatic.
Soil salinity has been measured using electrical conductivity for more than 100 years.
Units of Electrical Conductivity
The SI unit for electrical conductance is the Siemen, so electrical conductivity has units of S/m. Units used in older literature are mho/cm (mho is reciprocal ohm), which have the same value as S/cm. Soil electrical conductivities were typically reported in mmho/cm so 1 mmho/cm equals 1 mS/cm. Since SI discourages the use of submultiples in the denominator, this unit is changed to deciSiemen per meter (dS/m), which is numerically the same as mmho/cm or mS/cm. Occasionally, EC is reported as mS/m or µS/m. 1 dS/m is 100 mS/m or 105 µS/m.
Richards, L. A. (Ed.) 1954. Diagnosis and Improvement of Saline and Alkali Soils. USDA Agriculture Handbook 60, Washington D. C.
Rhoades, J. D. and J. Loveday. 1990. Salinity in irrigated agriculture. In Irrigation of Agricultural Crops. Agronomy Monograph 30:1089-1142. Americal Society of Agronomy, Madison, WI.
Get more information on applied environmental research in our