# Posts from the ‘Low Impact Development’ Category

## Chalk Talk: Intensive vs. Extensive Variables

Learn the difference between intensive and extensive variables and how they relate to soil water potential vs. soil water content in our new Chalk Talk whiteboard series. In this video series, Dr. Colin S. Campbell teaches basic principles of environmental biophysics and how they relate to measuring different parameters of the soil-plant-atmosphere continuum.

## Watch the video

To learn more about measuring water potential vs. water content read: Why soil moisture sensors can’t tell you everything.

Download the “Researcher’s complete guide to water potential”—>

Download the “Researcher’s complete guide to soil moisture”—>

## Video transcript

Hello, my name is Colin Campbell. I’m a senior research scientist here at METER group. And I teach a class on environmental biophysics. Today I wanted to talk about something we teach in the class: the difference between extensive and intensive variables. I’d like to do this with the goal of relating it to the difference between volumetric water content and water potential.

Here, I have a picture of a ship moving through the ice and some metal that’s been heated in a furnace. I think we would agree the ship has the highest amount of heat in it compared to this very small piece of metal. And if we placed that piece of metal onto the outside of the ship, despite the fact that there is more heat in the metal, we know the heat would not move from the high amount of heat (ship) to the low amount of heat (metal). It would actually move from the highest temperature to the lowest temperature. Why is that?

The reason is that heat moves because of temperature and not because of heat content or the amount of heat in something. Heat content defines an amount or an extent. And we generally term something that defines an extent or an amount as an extensive variable.An extensive variable depends broadly on the size of something or how much of something there is.

This differs for temperature. Temperature doesn’t depend on size. The temperature could be the same in a very small room or a very large room, but the amount of heat or heat content in those rooms would be quite different. When we describe temperature, we talk about intensity, which is why we call these types of variables intensive variables. This is because they don’t depend on size or amount.

Let’s talk about another example. Here’s your heating bill. Maybe it’s natural gas. What you’re paying for is the amount of heat you put into the house. But the question is, are you comfortable in the house? And from this bill, we can’t tell. Maybe you put in 200 heat units, whatever those might be. We can’t tell if that’s comfortable because we don’t know the size of the house or the type of insulation. All those things would influence whether you were comfortable.

Alternatively, if the temperature is 71 F that’s quite comfortable. That’s equivalent to about 22 degrees Celsius. So the intensive variable, temperature, is different than the extensive variable, heat content, that tells us how much heat we put in. And that’s important because at the end of the day, that leads to cost.

On this side, we don’t know how much we paid to keep it at 22 C because heat content doesn’t tell us anything about that. But the intensive variable temperature does tell us something about comfort. So both of these variables are critical to really understanding something about our comfort in the house.

Now let’s talk about the natural environment. Specifically, we’re going to talk about soils. We’ll start with the extensive variable. When we talk about water in soil, the extensive variable is, of course, water content. Water content defines the amount of water. Why would we care about water content? Well, for irrigation or a water balance.

The intensive variable is called water potential. What does water potential tell us? It tells us if soil water is available and also predicts water movement. If this soil had a water content of 25% VWC and another soil was at 20% VWC, would the water move from the higher water content to the lower water content? Well, that would be like our example of the ship and the heated piece of metal. We don’t know if it would move. It may move. And if the soil on either side was exactly the same, we might presume that it would move from the higher water content to the lower water content, but we actually don’t know. Because the water content is an extensive variable, it only tells us how much there is. It won’t tell us if it will move.

Now, if we knew that this soil water potential was -20 kPa and this soil water potential over here was -15 kPa, we would know something about where the water would move, and it would do something different than we might think. It would move from the higher water potential to the lower water potential against the gradient in water content, which is pretty interesting but nonetheless true. Water always moves from the highest water potential to the lowest water potential.

This helps us understand these variables in terms of plant comfort. We talked about the temperature being related to human comfort. We know at what temperatures we are most comfortable. With plants, we know exactly the same thing, and we always turn to the intensive variable, water potential, to define plant comfort.

For example, if we have an absolute scale like water potential for a particular plant, let’s say -15 kPa is the upper level for plant comfort, and -100 kPa is the lower level of comfort, we could keep our water potential in this range. And the plant would be happy all the time. Just like if we kept our temperature between 21 and 23 Celsius, that would be comfortable for humans. But of course, we humans are different. Some people think that temperature is warm, and some think it’s cold. And it’s the same for plants. So this isn’t a hard and fast rule. But we can’t say the same thing with water content. There’s no scale where we can say at 15% water content up to 25% water content you’ll have a happy plant That’s not true.If we know something about the soil, we can infer it. But soil is unique. And we’d have to derive this relationship between the water content and the water potential to know that.

So why would we ever think about using water content when we measure water in the soil? One reason is it’s the most familiar to people. And it’s the simplest to understand. It’s easy to understand an amount. But more importantly, when we talk about things like how much we’re going to irrigate, we might need to put on 10 millimeters of water to make the plants happy. And we’d need to measure that. Also if we want to know the fate of the water in the system, how much precipitation and irrigation we put on versus how much is moving down through the soil into the groundwater, that also relates to an amount.

But when we want to understand more about plant happiness or how water moves, it’s going to be this intensive variable, water potential that makes the biggest difference. And so with that, I’ll close. I’d love for you to go check out our website www.metergroup.com to learn a little bit more about these measurements in our knowledge base. And you’re also welcome to email me about this at colin.campbell@meter group.com.

## Just released: ATMOS 41 comparison testing data

Climate parameters such as precipitation, air temperature, and wind speed can change considerably across short distances in the natural environment. However, most weather observations either sacrifice spatial resolution for scientific accuracy or research-grade accuracy for spatial resolution.

ATMOS 41 all-in-one weather station

The ATMOS 41 represents an optimization of both. It was carefully engineered to maximize accuracy at a price point that allows for spatially distributed observations. Additionally, because many researchers need to avoid frequent maintenance and long setup times, the ATMOS 41 weather station was designed to reduce complexity and withstand long-term deployment in harsh environments. To eliminate breakage, it contains no moving parts, and it only requires recalibration every two years. Since all 14 measurements are combined in a single unit, it can be deployed quickly and with almost no effort. Its only requirement is to be mounted and leveled on top of a pole with an unobstructed view of the sky.

## Comparison testing and sensor-to-sensor variability data

METER released the ATMOS 41 in January 2017 after extensive development and testing with partnerships across the world, in Africa, Europe, and the US. We performed comparison testing with high-quality, research-grade non-METER sensors and conducted time-series testing for sensor-to-sensor variability.

## See performance data for the ATMOS 41 weather station

Download the “Researcher’s complete guide to soil moisture”—>

Download the “Researcher’s complete guide to water potential”—>

## Hydrology 301: What a Hydraulic Conductivity Curve Tells You & More

Hydraulic conductivity is the ability of a porous medium (soil for instance) to transmit water in saturated or nearly saturated conditions. It’s dependent on several factors: size distribution, roughness, tortuosity, shape, and degree of interconnection of water-conducting pores. A hydraulic conductivity curve tells you, at a given water potential, the ability of the soil to conduct water.

One factor that affects hydraulic conductivity is how strong the structure is in the soil you’re measuring.

For example, as the soil dries, what is the ability of water to go from the top of a sample [or soil layer in the field] to the bottom. These curves are used in modeling to illustrate or predict what will happen to water moving in a soil system during fluctuating moisture conditions. Researchers can combine hydraulic conductivity data from two laboratory instruments, the KSAT and the HYPROP, to produce a full hydraulic conductivity curve (Figure 1).

Figure 1. Example of hydraulic conductivity curves for three different soil types. The curves go from field saturation on the right to unsaturated hydraulic conductivity on the left.  They illustrate the difference between a well-structured clayey soil to a poorly structured clayey soil and the importance of structure to hydraulic conductivity especially at, or near, saturation.

In Hydrology 301, Leo Rivera, Research Scientist at METER, discusses hydraulic conductivity and the advantages and disadvantages of methods used to measure it.

Watch the webinar below.

Get more info on applied environmental research in our

Download the “Researcher’s complete guide to water potential”—>

Download the “Researcher’s complete guide to soil moisture”—>

## How to Protect your Soil Moisture Sensors from Lightning Surge

We occasionally see soil moisture sensors damaged by lightning.  Here’s what to do to protect them.

The secondary products of a lightning strike include electromagnetic pulses, electrostatic pulses, and earth current transients.

Surge suppression components typically perform their suppression function by temporarily short circuiting the voltage between two wires, several devices, or ground.

Electromagnetic pulses are created by the strong magnetic field that is formed by the short term current flow taking place in the lightning strike. With current flows as high as 510kA per microsecond, these currents create very large magnetic fields. These short-term magnetic fields then induce voltages onto wires and cables.

Electrostatic pulses are created by electrostatic fields that accompany a thunderstorm. Any cable suspended above the earth during a thunderstorm is immersed in the electrostatic field and will be electrically charged. Quick changes in the charges stored in both the clouds and earth take place whenever there is a lightning strike. The charge on the cable must now be discharged or neutralized. Unable to find a path to ground (earth), it breaks down insulation and component in its efforts to return to earth.

Earth current transients are the direct result of the neutralization process that immediately follows the end of lightning strike. Neutralization is accomplished by the movement or redistribution of charge along or near the earth’s surface from all the points where the charge had been initially induced to the point where the lightning strike has just terminated. Earth current transients create a shift in potential across a ground plan, often called a “ground bounce”.

Download the “Researcher’s complete guide to soil moisture”—>

Download the “Researcher’s complete guide to water potential”—>

## Soil Moisture Sensors: Which Installation Method is Best?

Patterns of water replenishment and use give rise to large spatial variations in soil moisture over the depth of the soil profile. Accurate measurements of profile water content are therefore the basis of any water budget study. When monitored accurately, profile measurements show the rates of water use, amounts of deep percolation, and amounts of water stored for plant use.

## How to avoid measurement errors

Three common challenges to making high-quality volumetric water content measurements are:

1. making sure the probe is installed in undisturbed soil,
2. minimizing disturbance to roots and biopores in the measurement volume, and
3. eliminating preferential water flow to, and around, the probe.

All dielectric probes are most sensitive at the surface of the probe. Any loss of contact between the probe and the soil or compaction of soil at the probe surface can result in large measurement errors. Water ponding on the surface and running in preferential paths down probe installation holes can also cause large measurement errors.

Installing soil moisture sensors will always involve some digging. How do you accurately sample the profile while disturbing the soil as little as possible?  Consider the pros and cons of five different profile sampling strategies.

## Preferential flow is a common issue with commercial profile probes

Profile probes are a one-stop solution for profile water content measurements. One probe installed in a single hole can give readings at many depths. Profile probes can work well, but proper installation can be tricky, and the tolerances are tight. It’s hard to drill a single, deep hole precisely enough to ensure contact along the entire surface of the probe. Backfilling to improve contact results in repacking and measurement errors. The profile probe is also especially susceptible to preferential-flow problems down the long surface of the access tube.  (NOTE: The new TEROS Borehole Installation Tool eliminates preferential flow and reduces site disturbance while allowing you to install sensors at depths you choose.)

## Trench installation is arduous

Installing sensors at different depths through the side wall of a trench is an easy and precise method, but the actual digging of the trench is a lot of work. This method puts the probes in undisturbed soil without packing or preferential water-flow problems, but because it involves excavation, it’s typically only used when the trench is dug for other reasons or when the soil is so stony or full of gravel that no other method will work. The excavated area should be filled and repacked to about the same density as the original soil to avoid undue edge effects.

Digging a trench is a lot of work.

## Augur side-wall installation is less work

Installing probes through the side wall of a single augur hole has many of the advantages of the trench method without the heavy equipment. This method was used by Bogena et al. with EC-5 probes. They made an apparatus to install probes at several depths simultaneously. As with trench installation, the hole should be filled and repacked to approximately the pre-sampling density to avoid edge effects.

An augered borehole disturbs the soil layers, but the relative size of the impact to the site is a fraction of what it would be with a trench installation. A trench may be about 60 to 90 cm long by 40 cm wide. A borehole installation performed using a small hand auger and the TEROS Borehole Installation Tool creates a hole only 10 cm in diameter—just 2-3% of the area of a trench. Because the scale of the site disturbance is minimized, fewer macropores, roots, and plants are disturbed, and the site can return to its natural state much faster. Additionally, when the installation tool is used inside a small borehole, good soil-to-sensor contact is ensured, and it is much easier to separate the horizon layers and repack to the correct soil density because there is less soil to separate.

## Multiple-hole installation protects against failures

Digging a separate access hole for each depth ensures that each probe is installed into undisturbed soil at the bottom of its own hole. As with all methods, take care to assure that there is no preferential water flow into the refilled augur holes, but a failure on a single hole doesn’t jeopardize all the data, as it would if all the measurements were made in a single hole.

The main drawback to this method is that a hole must be dug for each depth in the profile. The holes are small, however, so they are usually easy to dig.

## Single-hole installation is least desirable

It is possible to measure profile moisture by auguring a single hole, installing one sensor at the bottom, then repacking the hole, while installing sensors into the repacked soil at the desired depths as you go. However, because the repacked soil can have a different bulk density than it had in its undisturbed state and because the profile has been completely altered as the soil is excavated, mixed, and repacked, this is the least desirable of the methods discussed. Still, single-hole installation may be entirely satisfactory for some purposes. If the installation is allowed to equilibrate with the surrounding soil and roots are allowed to grow into the soil, relative changes in the disturbed soil should mirror those in the surroundings.

## Reference

Bogena, H. R., A. Weuthen, U. Rosenbaum, J. A. Huisman, and H. Vereecken. “SoilNet-A Zigbee-based soil moisture sensor network.” In AGU Fall Meeting Abstracts. 2007. Article link.

Read more soil moisture sensor installation tips.

Download the “Researcher’s complete guide to soil moisture”—>

Download the “Researcher’s complete guide to water potential”—>

## Soil Moisture Sensors: Why TDR VS. Capacitance May Be Missing the Point (Part 2)

Dr. Colin S. Campbell discusses whether TDR vs. capacitance (see part 1) is the right question, the challenges facing soil moisture sensor technology, and the correct questions to ask before investing in a sensor system.

It’s easy to overlook the obvious question: what is being measured?

## What are You Trying to Measure?

When considering which soil water content sensor will work best for any application, it’s easy to overlook the obvious question: what is being measured?  Time Domain Reflectometry (TDR) vs. capacitance is the right question for a researcher who is looking at the dielectric permittivity across a wide measurement frequency spectrum (called dielectric spectroscopy). There is important information in these data, like the ability to measure bulk density along with water content and electrical conductivity. If this is the desired measurement, currently only one technology will do: TDR. The reflectance of the electrical pulse that moves down the conducting rods contains a wide range of frequencies.  When digitized, these frequencies can be separated by fast fourier transform and analyzed for additional information.

The objective for the majority of scientists, however, is to simply monitor soil water content instantaneously or over time, with good accuracy. There are more options if this is the goal, yet there are still pitfalls to consider.

Considerable research has been devoted to determining which soil moisture sensors meet expectation.

## Each Technology Has Challenges

Why would a scientist pay \$100+ for a soil volumetric water content (VWC) sensor, when there are hundreds of soil moisture sensors online costing between \$5 and \$15? This is where knowing HOW water content is measured by a sensor is critical.

Most sensors on home and garden websites work based on electrical resistivity or conductivity. The principle is simple: more water will allow more electrons to flow. So conductivity will change with soil water content. But, while it’s possible to determine whether water content has changed with this method, absolute calibration is impossible to achieve as salts in the soil water will change as the water content changes. A careful reading of sensor specs will sometimes uncover the measurement method, but sometimes, price is the only indication.

Somewhere between dielectric spectroscopy and electrical resistance are the sensors that provide simple, accurate water content measurement. Considerable research has been devoted to determining which of these meet expectation, and the results suggest that Campbell Scientific, Delta-T, Stevens, Acclima, Sentek, and METER (formerly Decagon Devices), provide accurate sensors vetted by soil scientists. The real challenge is installing the sensors correctly and connecting them to a system that meets data-collection and analysis needs.

## Installation Techniques Affect Accuracy

Studies show there is a difference between mid-priced sensor accuracy when tested in laboratory conditions. But, in the field, sensor accuracy is shown to be similar for all good quality probes, and all sensors benefit from site-specific soil calibration. Why? The reason is associated with the principle upon which they function. The electromagnetic field these sensors produce falls off exponentially with distance from the sensor surface because the majority of the field is near the electrodes. So, in the lab, where test solutions form easily around sensor rods, there are differences in probe performance.  In a natural medium like soil, air gaps, rocks, and other detritus reduce the electrode-to-soil contact and tend to reduce sensor to sensor differences. Thus, picking an accurate sensor is important, but a high-quality installation is even more critical.

Improper installation is the largest barrier to accuracy.

## Which Capacitance Sensor Works Best?

Sensor choice should be based on how sensors will be installed, the nature of the research site, and the intended collection method. Some researchers prefer a profile sensor, which allows instruments to be placed at multiple depths in a single hole. This may facilitate fast installation, but air gaps in the auger pilot hole can occur, especially in rocky soils. Fixing this problem requires filling the hole with a slurry, resulting in disturbed soil measurements. Still, profile sensor installation must be evaluated against the typical method of digging a pit and installing sensors into a sidewall. This method is time consuming and makes it more difficult to retrieve sensors.

New technology that allows sensor installation in the side of a 10 cm borehole may give the best of both worlds, but still requires backfill and has the challenge of probe removal at the end of the experiment.

The research site must also be a consideration. If the installation is close to main power or easily reached with batteries and solar panels, your options are open: all sensors will work. But, if the site is remote, picking a sensor and logging system with low power requirements will save time hauling in solar panels or the frustration of data loggers running out of batteries.

Often times it comes down to convenience.

## Data Loggers Can Be a Limitation

Many manufacturers design data loggers that only connect to the sensors they make. This can cause problems if the logging system doesn’t meet site needs. All manufacturers mentioned above have sensors that will connect to general data loggers such as Campbell Scientific’s CR series. It often comes down to convenience: the types of sensor needed to monitor a site, the resources needed to collect and analyze the data, and site maintenance. Cost is an issue too, as sensors range from \$100 to more than \$3000.

## Successfully Measure Water Content

The challenge of setting up and monitoring soil water content is not trivial, with many choices and little explanation of how each type of sensor will affect the final results. There are a wealth of papers that review the critical performance aspects of all the sensors discussed, and we encourage you to read them. But, if soil water content is the goal, using one of the sensors from the manufacturers named above, a careful installation, and a soil-specific calibration, will ensure a successful, accurate water content measurement.

For an in-depth comparison of TDR versus capacitance technology, read: Dielectric Probes Vs. Time Domain Reflectometers

## Watch the webinar

In this webinar, Dr. Colin Campbell discusses the details regarding different ways to measure soil moisture and the theory behind the measurements.  In addition, he provides examples of field research and what technology might apply in each situation. The measurement methods covered are gravimetric sampling, dielectric methods including TDR and FDR/capacitance, neutron probe, and dual needle heat pulse.

Download the “Researcher’s complete guide to soil moisture”—>

## Soil Moisture Sensors: Why TDR vs. Capacitance May Be Missing the Point

Time Domain Reflectometry (TDR) vs. capacitance is a common question for scientists who want to measure volumetric water content (VWC) of soil, but is it the right question?  Dr. Colin S. Campbell, soil scientist, explains some of the history and technology behind TDR vs. capacitance and the most important questions scientists need to ask before investing in a sensor system.

TDR began as a technology the power industry used to determine the distance to a break in broken power lines.

## Clarke Topp

In the late 1970s, Clarke Topp and two colleagues began working with a technology the power industry used to determine the distance to a break in broken power lines.  Time Domain Reflectometers (TDR) generated a voltage pulse which traveled down a cable, reflected from the end, and returned to the transmitter. The time required for the pulse to travel to the end of the cable directed repair crews to the correct trouble spot. The travel time depended on the distance to the break where the voltage was reflected, but also on the dielectric constant of the cable environment.  Topp realized that water has a high dielectric constant (80) compared to soil minerals (4) and air (1).  If bare conductors were buried in soil and the travel time measured with the TDR, he could determine the dielectric constant of the soil, and from that, its water content.  He was thus able to correlate the time it took for an electromagnetic pulse to travel the length of steel sensor rods inserted into the soil to volumetric water content. Despite his colleagues’ skepticism, he proved that the measurement was consistent for several soil types.

TDR sensors consume a lot of power. They may require solar panels and larger batteries for permanent installations.

## TDR Technology is Accurate, but Costly

In the years since Topp et al.’s (1980) seminal paper, TDR probes have proven to be accurate for measuring water content in many soils. So why doesn’t everyone use them? The main reason is that these systems are expensive, limiting the number of measurements that can be made across a field. In addition, TDR systems can be complex, and setting them up and maintaining them can be difficult.  Finally, TDR sensors consume a lot of power.  They may require solar panels and larger batteries for permanent installations. Still, TDR has great qualities that make these types of sensors a good choice.  For one thing, the reading is almost independent of electrical conductivity (EC) until the soil becomes salty enough to absorb the reflection.  For another, the probes themselves contain no electronics and are therefore good for long-term monitoring installations since the electronics are not buried and can be accessed for servicing, as needed.  Probes can be multiplexed, so several relatively inexpensive probes can be read by one set of expensive electronics, reducing cost for installations requiring multiple probes.

Many modern capacitance sensors use high frequencies to minimize effects of soil salinity on readings.

## Advances in Electronics Enable Capacitance Technology

Dielectric constant of soil can also be measured by making the soil the dielectric in a capacitor.  One could use parallel plates, as in a conventional capacitor, but the measurement can also be made in the fringe field around steel sensor rods, similar to those used for TDR.  The fact that capacitance of soil varies with water content was known well before Topp and colleagues did their experiments with TDR.  So, why did the first attempt at capacitance technology fail, while TDR technology succeeded? It all comes down to the frequency at which the measurements are made.  The voltage pulse used for TDR has a very fast rise time.  It contains a range of frequencies, but the main ones are around 500 MHz to 1 GHz.  At this high frequency, the salinity of the soil does not affect the measurement in soils capable of growing most plants.

Like TDR, capacitance sensors use a voltage source to produce an electromagnetic field between metal electrodes (usually stainless steel), but instead of a pulse traveling down the rods, positive and negative charges are briefly applied to them. The charge stored is measured and related to volumetric water content. Scientists soon realized that how quickly the electromagnetic field was charged and discharged was critical to success.  Low frequencies led to large soil salinity effects on the readings.  This new understanding, combined with advances in the speed of electronics, meant the original capacitance approach could be resurrected. Many modern capacitance sensors use high frequencies to minimize effects of soil salinity on readings.

NASA used capacitance technology to measure water content on Mars.

## Capacitance Today is Highly Accurate

With this frequency increase, most capacitance sensors available on the market show good accuracy. In addition, the circuitry in them can be designed to resolve extremely small changes in volumetric water content, so much so, that NASA used capacitance technology to measure water content on Mars. Capacitance sensors are lower cost because they don’t require a lot of circuitry, allowing more measurements per dollar. Like TDR, capacitance sensors are reasonably easy to install. The measurement prongs tend to be shorter than TDR probes so they can be less difficult to insert into a hole. Capacitance sensors also tend to have lower energy requirements and may last for years in the field powered by a small battery pack in a data logger.

In two weeks: Learn about challenges facing both types of technology and why the question of TDR vs. Capacitance may not be the right question.

## Watch the webinar

In this webinar, Dr. Colin Campbell discusses the details regarding different ways to measure soil moisture and the theory behind the measurements.  In addition, he provides examples of field research and what technology might apply in each situation. The measurement methods covered are gravimetric sampling, dielectric methods including TDR and FDR/capacitance, neutron probe, and dual needle heat pulse.

Get more information on applied environmental research in our

Download the “Researcher’s complete guide to soil moisture”—>

## New Infiltrometer Helps City of Pittsburgh Limit Traditional Stormwater Infrastructure

Though difficult and expensive to restore, the brick-paved streets that still exist in some Pennsylvania neighborhoods are a treasure worth preserving, according to the City of Pittsburgh. Dellrose Street, an aging, 900 ft. long, brick road, was in need of repair, but the city of Pittsburgh wanted to limit traditional stormwater infrastructure, such as pipes and catch basins.

Dellrose Street permeable paver system

To save the aesthetics of the neighborhood, they hired ms consultants, inc. to design a permeable paver solution for controlling stormwater runoff volumes and peak runoff rates that would traditionally be routed off-site via storm sewers.  Jason Borne, a stormwater engineer for ms consultants who worked on the project says, “What we try to do is understand the in situ infiltration potential of the subsoils to determine the most efficient natural processes for attenuating flows; either through infiltrating excess water volume back into the soil or through slow-release off-site.”  He used the SATURO Infiltrometer to get an idea of how urban fill material would infiltrate water.

## Green Infrastructure Aids Natural Infiltration

As Borne and his team investigated what they could do to slow down the runoff, they decided permeable pavers would be a viable solution.  He says, “There’s not much you can do once you put in a hardened surface like a pavement.  Traditional pavement surfaces accelerate the runoff which requires catch basins and large diameter pipes to carry the runoff off-site. We were interested in investigating what some of the urban subsoils or urban fill would allow us to do from an infiltration perspective.  As we started looking at some of these subsoils, we decided a permeable paver system would be ideal for this particular street.”

Subsurface flow barrier installation

## Infiltrometers Determine Natural Infiltration Potential

Once the water flowed into the aggregate, the team began to figure out ways to slow it down and promote infiltration.  Borne says, “Basically we came up with a tiered subsurface flow barrier system.  We had about 60 concrete flow barriers across the subgrade within the aggregate base of the road. We needed so many because the longitudinal slope of the road was fairly significant. Behind each of these barriers we stored a portion of the stormwater that would typically run off the site.  The ideal was to remove the stored water through infiltration—to get it down to the subgrade and away, so we used infiltrometers to help us establish where we could maximize infiltration and where we might need to rely on other management methods.”

## A Need for Faster Test Times Inspires a Comparison

Borne says that USDA soil surveys are too generalized for green infrastructure applications in urban areas and only give crude approximations of the soil hydraulic conductivity. Understanding the best way to promote natural infiltration requires a very specific infiltration rate or hydraulic conductivity for the location of interest.  He says, “The goal is to excavate down to the desired elevation before construction and find out, through some kind of device what the infiltration potential of the subsoil is.  Typically we use a double ring infiltrometer, but it’s a very manual device. We’re constantly refilling water, and it requires us to be on-site and attentive to what’s happening.  We can’t really multitask, especially in areas of decently infiltrating soils where the device might run out of water in 30 minutes or less. So, in the interest of saving water and time, we used the automated SATURO infiltrometer and the manual double ring infiltrometer concurrently for comparison purposes.”

Next week:  Find out how the two infiltrometers compared.

Download the “Researcher’s complete guide to soil moisture”—>

Get more information on applied environmental research in our

## Top Five Blog Posts in 2016

In case you missed them the first time around, here are the most popular Environmental Biophysics.org blog posts in 2016.

## Lysimeters Determine if Human Waste Composting can be More Efficient

In Haiti, untreated human waste contaminating urban areas and water sources has led to widespread waterborne illness.  Sustainable Organic Integrated Livelihoods (SOIL) has been working to turn human waste into a resource for nutrient management by turning solid waste into compost.  Read more

## Estimating Relative Humidity in Soil: How to Stop Doing it Wrong

Estimating the relative humidity in soil?  Most people do it wrong…every time.  Dr. Gaylon S. Campbell shares a lesson on how to correctly estimate soil relative humidity from his new book, Soil Physics with Python, which he recently co-authored with Dr. Marco Bittelli.  Read more.

## How Many Soil Moisture Sensors Do You Need?

“How many soil moisture sensors do I need?” is a question that we get from time to time. Fortunately, this is a topic that has received substantial attention by the research community over the past several years. So, we decided to consult the recent literature for insights. Here is what we learned.

## Data loggers: To Bury, or Not To Bury

Globally, the number one reason for data loggers to fail is flooding. Yet, scientists continue to try to find ways to bury their data loggers to avoid constantly removing them for cultivation, spraying, and harvest.  Chris Chambers, head of Sales and Support at Decagon Devices always advises against it. Read more

## Founders of Environmental Biophysics:  Champ Tanner

Image: http://soils.wisc.edu/people/history/champ-tanner/

We interviewed Gaylon Campbell, Ph.D. about his association with one of the founders of environmental biophysics, Champ Tanner.  Read more

## Do the Standards for Field Capacity and Permanent Wilting Point Need to Be Reexamined?

We asked scientist, Dr. Gaylon S. Campbell, which scientific idea he thinks impedes progress.  Here’s what he had to say about the standards for field capacity and permanent wilting point.  Read more

## Environmental Biophysics Lectures

During a recent semester at Washington State University, a film crew recorded all of the lectures given in the Environmental Biophysics course. The videos from each Environmental Biophysics lecture are posted here for your viewing and educational pleasure.  Read more

## Soil Moisture Sensors In a Tree?

Soil moisture sensors belong in the soil. Unless, of course, you are feeling creative, curious, or bored. Then maybe the crazy idea strikes you that if soil moisture sensors measure water content in the soil, why couldn’t they be used to measure water content in a tree?  Read more

Download the “Researcher’s complete guide to soil moisture”—>

Get more information on applied environmental research in our

## Data Logger Dilemma: To Bury, or Not to Bury—An Update

Recently, we wrote about scientists who were burying their data loggers (read it here).  Radu Carcoana, research specialist and Dr. Aaron Daigh, assistant professor at North Dakota State University, used paint cans to completely seal their data loggers before burying them in the fall of 2015.

Paint can setup for buried data logger.

They drilled ports for the sensor cables, sealed them up, and when they needed to collect data, they dug up the cans, retrieved the instruments, and downloaded the data in a minute or less.

Here Radu gives an update of what happened when he dug up his buried instruments in the spring.

## Results of the Paint Can Experiment

In May of this year, we dug up eighteen units (one data logger and four soil moisture sensors per unit) left in the field since November 2015—over six months.

Did moisture get into the paint cans? —We found only three cans with water in them, purely due to installation techniques used for that specific unit. The other fifteen units were bone dry, although total precipitation for the month of April only amounted to 3.63 inches, plus the snow melt.

How was data recording and recovery? —For six months, every 30 minutes the soil moisture sensors took readings, the data logger recorded, and we retrieved all of the data, complete and unaltered.

Only three cans with water in them, due to installation techniques.

What about power consumption? The batteries were good —over 90% did not need replacement. The power budget provided by five AA batteries was more than enough for reading four soil moisture sensors at 30-minute intervals.

## What Happens Now?

In the spring of this year, we installed 18 more units in the third farm field, right after planting soya. We now have 36 individual units (~\$1,000 value each unit) buried in the ground in the middle of a field planted with corn or soybean, since the beginning of May.

On October 13-14 (after 5 months), we accessed the first twelve units (Farm A). All 30 minutes of data was read, recorded, and downloaded (since May).  The batteries and the other accessories were replaced, and then we sealed and reburied the cans. Only one unit out of twelve had an issue and was replaced: the battery exploded in the can (editor’s note: battery explosion is usually caused by a manufacturing defect and the risk can be lessened by purchasing higher quality batteries, although all types are susceptible to some degree).  Since battery leakage will often corrode everything the acid touches, the data logger had to be sent back for repair and there may be partial data loss. The other 24 units (Farm B and C) will be accessed next week, weather permitting.

Over 90% of batteries did not need replacement.

## Is the Paint Can Method Worth it?

We will continue to monitor and retrieve the data from the buried data loggers (We don’t use data loggers suited for wireless communication, because several factors guided us not to). The paint can system works very well if the installation is done correctly, with great attention to detail, and it costs only \$2.00/can. However, there are improvements that could be made in order to have this method become a standard in soil research. For instance, though we are still using paint cans and other common materials, advancements in the design of waterproof containers and sturdiness would be a huge step forward. This is just a well thought out concept – a prototype. It proves that burying electronics for a longer period of time can be done if properly executed.

Note:  METER’s (formerly Decagon) official position is that you should never bury your data logger.  But we couldn’t resist sharing a few stories of scientists who have figured out some innovative methods which may or may not be successful, if tried at other sites.

Download the “Researcher’s complete guide to soil moisture”—>

Get more information on applied environmental research in our