The PHYTOS 31 Leaf Wetness Sensor was designed to measure the presence and duration of water on leaf surfaces. However, Dr. Bruce Bugbee, professor of Crop Physiology at Utah State University, noticed that his leaf wetness sensor revealed interesting phenomena associated with some precipitation events. Here is what he observed on a recent day at the USU Environmental Observatory in Logan, Utah
“Recent data from our weather station provided two examples of the offset in measurement associated with tipping bucket rain gauges. It started raining on campus last night at exactly 20:00 hours, as indicated by the response of the leaf wetness sensor (Figure 1). The first 0.1 mm tip of the rain gauge occurred about 25 minutes later (Figure 2). The resolution for most high-quality tipping bucket rain gauges is listed as 0.1 mm, but this is not the resolution for the first 0.1 mm of rain.